Far infrared Therapy House common Check.
(1) short-circuit equipment or fuse blown: A, check the short-circuit current is consistent with its requirements; B, to determine whether the same circuit too much electrical overload; C, replace the same specifications fuse.
(2) the light does not shine: A, to determine the connection of the roof-related plug; B, replace the lamp.
(3) After the spectrum board is not heated: A, open the lower right front panel power box wood cover (from the inside), check the base of the wire plug and the relevant socket connection; B, remove the sitting plate, check the power hand Connection to socket.
(4) There is no frequency spectrum plate temperature before and after the board temperature: This is determined by design requirements, not the fault phenomenon, sitting board spectrum board is working properly.
(5) spectrum therapy device power: A, check the main power plug and cable outlet; B, open the lower right front of the power box wood cover (from the inside). Ensure that the plug at the end of the main power cord is properly connected.